Shenyang Best Cleaning Equipment CO.,LTD.
Contact: manager li
Email address: firstname.lastname@example.org
Zip code: 110005
Address: 12th floor,jizhi building,no.150 nanjing south street,heping district,shenyang city
1. After turning on the power supply, the indicator light does not light up, the industrial oxygen generator does not operate, and there is an audible and visual alarm.
In this case, first of all, check whether the plug of the power supply is correctly inserted into the power socket. If not, plug it tightly and then check whether the household oxygen generator is turned on. If the power socket is not powered, you can use a measuring pen to check. If there is no such phenomenon, consider whether the household oxygen generator will be overloaded This constitutes the intelligent distinction of the medical oxygen generator and protects itself. To clean up this element, press the reset button of the overload protector of the domestic oxygen generator. If there is still no doubt, it can be the loosening of the wiring on the oxygen generator. All the wiring can be reset from the beginning and then inspected. If there is still no result, it can be the internal wiring question, which needs to be checked Return to the factory for repair, the user should contact the seller in time at this time, and be disposed in time for later use.
2. After turning on the power, the indicator light is on, but the industrial oxygen generator is still not working.
This situation can be the operation with the thermal protector of the air compressor. If the household oxygen generator will stop after 45 minutes of starting, it is urgent to contact the seller for repair. Or the air inlet or air outlet is blocked. Clean the filter properly, and check whether there is any obstruction at the air inlet and air outlet to the air inlet and oxygen outlet, and whether there is any substance that hinders the heat dissipation of the body. If none of the above conditions exist, it can be affected by room temperature. If the indoor environment temperature is lower than 5 ℃, it needs to be heated to ensure normal operation.
Shenyang Best Cleaning equipment co., LTD.
Address: 12th floor,jizhi building,no.150 nanjing
south street,heping district,shenyang city
Official Qr Code